|کد مقاله||کد نشریه||سال انتشار||مقاله انگلیسی||ترجمه فارسی||نسخه تمام متن|
|1880632||1533419||2014||4 صفحه PDF||سفارش دهید||دانلود رایگان|
• Dosimetry using OSL of BeO.
• Environmental dosimeter with modified personal dosimeter and linear algorithm.
• Use of doped BeO chips for better energy response.
In the course of this work, the possibility of the measurement of ambient dose equivalent H*(10) with the BeOSL dosimetry system was evaluated. Calculations for the energy response of the 2-element BeOSL dosimeter for irradiation with H*(10) were performed. The response doesn't fulfil the requirements of IEC 62387-1. Especially the response for photon energies of 60–100 keV is to low. It is possible to correct this under response using a modified BeOSL 2-element dosimeter and a linear algorithm. So the national requirements for a H*(10) dosimeter in Germany can be fulfilled. An incidence angle independent measurement is not possible because for several angles of incidences (>60°) the filters of the 2-element dosimeter doesn't shield the correct element. Another material which is more suitable for the H*(10) measurement was tested. So the doping of BeO with Lanthanum leads to an enhanced energy response for measurement of H*(10). Furthermore a higher OSL sensitivity was found for this material. Further tests on the influence of Lanthanum concentration on the dosimetric properties are necessary.
Journal: Radiation Measurements - Volume 71, December 2014, Pages 438–441