|کد مقاله||کد نشریه||سال انتشار||مقاله انگلیسی||ترجمه فارسی||نسخه تمام متن|
|875642||910789||2016||5 صفحه PDF||سفارش دهید||دانلود رایگان|
این مقاله ISI می تواند منبع ارزشمندی برای تولید محتوا باشد.
- تولید محتوا برای سایت و وبلاگ
- تولید محتوا برای کتاب
- تولید محتوا برای نشریات و روزنامه ها
پایگاه «دانشیاری» آمادگی دارد با همکاری مجموعه «شهر محتوا» با استفاده از این مقاله علمی، برای شما به زبان فارسی، تولید محتوا نماید.
• Ultrasonic tip vibrations depend on tip shape, design and generator power.
• Water running over the tip and loads suppress tip vibrations.
• Clinical loads with high power settings make the tips behave unpredictably.
• Low loads yield a steady decrease of tip vibration with increasing load.
The objective of this work was to investigate the performance characteristics of a piezoelectric ultrasonic dental scaler using scanning laser vibrometry. The vibration characteristics of three standard piezoelectric tips were assessed with scanning laser vibrometry under various conditions: unconstrained, under a stream of flowing water, in a water tank, as well as subjected to loads to simulate clinical conditions. Subsequently, the tips were used to disrupt an in-vitro biofilm model of dental plaque, developed using a non-pathogenic Gram-negative species of Serratia (NCIMB40259).The laser vibrometry data showed that the oscillation pattern of the ultrasonic tip depends primarily on its shape and design, as well as on the generator power. Thin tips and high power settings induce the highest vibrations. Water irrigation of the tip and loads influence the tip performance by diminishing its vibration, while water volume increases it.Serratia biofilm was disrupted by the cavitation bubbles occurring around the scaler tip. The most effective biofilm removal occurred with the thinner tip.Understanding how the ultrasonic tip oscillates when in use and how it removes dental plaque is essential for gaining more knowledge regarding the cleaning mechanisms of the ultrasonic system. Cavitation may be used to remove plaque and calculus without a mechanical contact between the dental tip and the teeth. Better knowledge would enable dental specialists to understand and improve their techniques during routine cleaning of teeth. It will also lead to improving tip design and to the production of more effective instruments for clinical use.
Journal: Medical Engineering & Physics - Volume 38, Issue 2, February 2016, Pages 199–203